Chillers / Circulators Haake F3-C Recirculating Heated Chiller Bath Haake F3C Refrigerated Circulator Haake F3-C Refrigerated Circulator. Find great deals for Haake F3 K Circulating Water Bath Chiller. Shop with confidence on eBay!. Find great deals for Haake F3-c Refrigerated Circulator V 50/60 Hz Great Shape. Shop with confidence on eBay!.
|Published (Last):||18 July 2010|
|PDF File Size:||15.68 Mb|
|ePub File Size:||9.49 Mb|
|Price:||Free* [*Free Regsitration Required]|
Image not available Photos not available for this variation.
Back to home page. Temperature The display value of the digital thermometer 6 corresponds to an analog voltage which is fed to the pins 5 and 1 neutral of socket 16 2. The bridge with F3 haakd assembled is placed onto the rear end of the refrigerated bath and fastened with two screws at A on the cover plate. If the unit is to cut off at an approximate value only, the excess temperature limiter 9 is turned somewhat in the other direction to the right after the cut-off value has been reached.
High operating temperatures will lead to high temperatures on c3 hose surface and even more so at the metal nozzles. Pressure fluctuations of the public water supply may hamper the temperature accuracy.
At the display the temperature and fluctuation of the external sensor are indicated, which might deviate considerably from the conditions within the circulator bath. Getting notified via SMS Haaek Message will require you to verify your cell phone via the profile page in your account. Set up the unit and allow a period of rest v3 24 hours.
Rotary Evaporator with 20L sample flask N-4000・N-4000H・N-4000W
This necessitates the following: When the temperature display flashes the set temperature exceeds the allowed operating temperature of the temp, control unit combination: This is a desired effect.
See haaake items More The counter piece for attaching them is illustrated to the left. There are 1 items available. If objects are to be temperature controlled in the internal bath only, the pump nozzles A 18 and 25 can be closed with a plug piece D attached with a coupling nut E.
Haame from these annoying effects the liquid level has to be constantly replenished. Hoses might become brittle after prolonged use or they may get very soft. This, in turn, causes the velocity of flow to increase to be noticed as a louder naake noise of the passing airthis however prevents inadmissible pressure conditions in the cooling circuit.
Electrical Connections At the rear panel of the F3 there are three sockets for connecting auxiliary equipment. See all condition definitions – opens in a new window or tab.
View auctioneer’s contact info Hot thermal liquid should not be drained! Range with additional cooling Cooling Coil Temp. Buyers must register prior to bidding at any auction sale. The liquid will drain out in a slight arc!
Used HAAKE F3 CH Water Bath For Sale – DOTmed Listing #
See terms – opens in a new window or tab. C Water Bath Model No: Control Sensor Excess Temp. Pin 2 “Neutral” and 6: The above table is an extract. Pt Sensor Connect, f. Effective Temperature Range Setting Range: For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab.
Connecting the Hoses Hoses are used to connect the pump with an external vessel. The haaek device must allow a reversal of the switching poles.
I Only personnel with the particular know how haaks be allowed to carry out repairs. This instruction manual describes how this circulator is to be used.
Haake F3-c Refrigerated Circulator V 50/60 Hz Great Shape | eBay
In order to avoid that the overtemperature sensor is triggered, it should be verified that The headings contained herein are t3 convenience only and shall not limit or alter the Terms hereof. Webcast bidders will be charged a Changes made with 3 can be noticed at the digital display. When certain conditions make draining necessary, please act safety conscious: Report item – opens in a new window or tab. Back to home page Return haakw top.
With a clamp in the pressure hose the amount of the circulating liquid is balanced with that of the amount floating back. Setting up 9 3.
Switch 5 to “Off” not depressedSwitch 28 to “extern” The external sensor has to be balanced as the length of the cable influences measuring results. Pin 2 “Neutral” and 5: This will prevent the water from gelling freezing in the area of the evaporating coil which is the coldest part of the compressor’s cooling circuit. When working with oil below room temperature: For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab This amount includes applicable customs duties, taxes, brokerage and other fees.
Visit your account to find out Synthetic thermal liquid with a medium life span some months and little smell annoyance.